Rare difficulties with I2C user bus data Transfer upon
setting attenuators or reading I2C devices.
Fix: removal of capacitor C48 (by trained personnel)
Shipments from October 10th 2016 onwards will have C48 removed.
Temporary fix without Hardware modification:
Set DACA output to maximum output (3.3V)
DWC8300 R10 cards are not affected.